What is the O2 measuring range of the sensor and the sensitivity of the sensor?
The measuring range on our model is 0.0 - 100% oxygen gas.
The accuracy is @ 59 degrees (F) to 104 degrees (F)is + / - 1% of full scale at constant temperature and pressure when calibrated at full scale and = / - 3% actual oxygen level over full operating temperature.
The Resolution s 0.1 percent Oxygen. The Display Range is 0.0-99.9 percent Oxygen. The Accuracy and Linearity is 1% of full scale at constant temperature, R.H & pressure when calibrated at full scale ?? 3% actual oxygen temperature range.